Search by Keyword
In the flow cell there are two different surfaces, top surface which is made of glass slides, and the bottom surface which is made of plastic. So my question is the bottom surface also as smooth as the top surface, or is it rough to enhance the attachment of the bacteria? If I grow the biofilms on the top surface instead of bottom surface by flipping the system, will the biofilms will be same or not?
Most Biofilms are grown on the top surface (or underside of the glass slide). This allows for slightly better observation of the biofilm via microscope. The bottom surface will also grow most Biofilms but it is slightly more abrasive than the top surface. To grow on top surface, the bacteria is introduced while the cell is turned upside down (glass surface facing downward) and allowed to sit still with no media flowing for maybe an hour or more to allow attachment and growth to begin.
Can flow cells be heated?
Yes, flow cells can be heated. For most biofilms to grow, the recommended temperature is from mid 30° to 42° C. What limits the temperature range for experiments are the different strengths and expansion coefficients between glass and plastic and the fact that at elevated (i.e. higher than 42°) temperatures the glass and plastic begin to upset the bonding glue which holds them together.
What is a recommended Flow Rate for the culture media through a flow cell?
There really isn’t a “recommended” flow rate. Different Biofilm cultures can require different flow rates of culture media. A “Suggested” flow rate or starting point is around 5ml per/hr. To calculate this flow rate there are several things that must be taken into consideration, such as: Tubing I.D. the specifications of the pump being used (the number of pump rollers affects the flow rate, see pump manual) and the number of channels in the flow cell (3-channel VS convertible).
What is the I.D. (Inner dimension) of the inlet and outlet tubing on the 3-channel flow cell?
The I.D. of the inlet and outlet tubing is 1/16”. This tubing is PVC material.
What is the overall length of the Flow cell assembly?
The overall length of the complete flow cell assembly is 91”.
What type of glue is used on the tube fittings found on the culture media bottle caps?
The glue used is a 3M product. It is a 2-part Epoxy that is rated for 162° C which should allow for autoclaving of the entire bottle and cap without issues.
Submit a Review...
Note: All prices in Australian Dollars